|Frederick H. Silver and Lisa L. Silver|
|Corresponding Author: Frederick H. Silver, Department of Pathology and Laboratory Medicine, Robert Wood Johnson Medical School, Rutgers, the State University of New Jersey, Piscataway, NJ 08854.|
|Received: October 24, 2017; Accepted: October 27, 2017; Published: October 29, 2017;|
|Citation: Silver F H & Silver L L, (2017) Non-Invasive Viscoelastic Behavior of Human Skin and Decellularized Dermis Using Vibrational OCT, 3(3): 174-179.|
|Copyrights: ©2017 Silver F H & Silver L L. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.|
The need to measure the mechanical properties of skin, scar, tumors, extracellular matrices, and wound tissue has been a goal of researchers since the 1970s. A variety of methods have been used to evaluate the mechanical properties of tissues over the last 40 years including uniaxial and biaxial tensile testing, indentation and rotational tests, ultrasound elastography (UE), optical cohesion tomography (OCT), optical cohesion elastography (OCE), and vibrational analysis combined with OCT.
We have developed a test using vibrational OCT to image and measure the moduli of the components of skin and scar tissue non-invasively and non-destructively. Using images generated by OCT, maps of the modulus as a function of position can be generated that will be useful in marking the margins of scars, tumors as well as to evaluate the effects of cosmetic treatments to the skin.
Key Words: Collagen, Extracellular matrix, Optical coherence tomography, Vibrational analysis, Skin, Scar, Tumor, Tumor margins
Measurement of the Modulus of Elastic Tissue and Collagen Fibers in Skin and Decellularized Dermis
Since the properties of skin are time-dependent and also depend on the rate of deformation of the sample, it is important to separate the elastic properties from the viscous properties . The model used in most of our studies was decellularized dermis since the mechanical properties of this material are similar to normal skin and decellularized dermis can be freeze dried and stored at room temperature in a sterilized form [3,5-8]. In addition it is also composed primarily of collagen fibers with some elastic tissue still present simplifying interpretation of the mechanical properties [3,5-8].
The non-linearity of the stress-strain curves was dealt with by dividing the stress-strain curves into elastic and viscous components [3,5-7, 9,10] (Figure 1). The elastic component was measured as the stress at equilibrium in an incremental stress-strain experiment [3,10]. This test was conducted by the sequential addition of loading increments, followed by a relaxation period between each loading step [3,10] (Figure 1). The elastic stress-strain curve was broken into a linear low modulus region and a linear high modulus region [3, 5-7]. Equilibrium stress-strain curves were then used to calculate the elastic modulus as the tangent to the curve which turned out to be about 0.5 MPa and 15 MPa, respectively for the low and high strain regions [3,5,10]. The low strain region modulus was similar to that reported for elastic tissue and the high strain modulus was similar to that reported for collagen fibers [7,10]. The major problem with this approach was that the time required for the stress to reach equilibrium in the skin and in decellularized dermis was as long as 24 hrs and the test resulted in destruction of the material.
The modulus of a number of ECMs was tested using this approach and the results indicated that the elastic modulus was independent of strain-rate for strain rates up to 10,000% per minute . When the slope of the elastic stress-strain curve at high strains is divided by the collagen content of the tissue, the fraction of collagen aligned with the tensile direction, and the change in strain after the collagen fibers of the tissue are stretched in tension, the resultant modulus was calculated to be between 4.0 and 7.0 GPa, values similar to those reported for the stiffness of the collagen molecule [10,11]. The slope of the viscous stress-strain curve reflected the length of the collagen fibrils and the viscous sliding of collagen fibrils by each other during tensile deformation .
Results of theoretical modeling studies suggested that the high strain elastic modulus was consistent with stretching of the collagen triple helical regions devoid of proline and hydroxyproline . Therefore, the stiffness or modulus at high strains reflects the resistance to stretching of the collagenous components of skin and is a useful parameter to evaluate skin properties and their changes during wound healing and disease processes. Results of studies on hypertrophic scar tissue have shown that the major difference between scar and normal skin mechanical properties lies in the inability of scar tissue to reorient under an applied load in the same manner as does normal skin when stretched in tension . However, since this approach required long time intervals and destruction of the tissue it would not be considered useful in studying skin mechanics in vivo. Therefore, our efforts became focused on development of a non-invasive and non-destructive test to measure the mechanical properties of skin in vivo.
Development of Vibrational Optical Cohesion Tomography (OCT)
The concept of development of a test to measure the mechanical properties of skin came from the observation that when one vibrates a bowl of soft material like jello with a hard inclusion, the jello vibrates at a higher frequency than the inclusion. If one were able to measure the frequency of vibration of the jello and inclusion separately then the frequency of vibration of each material should be related to the material stiffness. It turned out that a technique termed OCT created images of a substrate by comparing a beam of light with part of the beam that is reflected off a substrate . If one measures the displacement of each substrate at varying frequencies during vibration it is possible to calculate the modulus of each the material. Figure 2 shows a plot of weighted displacement versus frequency for decellularized dermis showing one resonant frequency (frequency of maximum displacement) for collagen in decellularized human dermis.
Recently, we have reported the use of vibrational analysis and OCT to characterize the mechanical behavior of decellularized dermis, pig skin, bovine cartilage and subchondral bone [5,6]. Our results indicate that Poisson’s ratio for decellularized dermis ranges between 0.38 and 0.63, values that are significantly different than 0.5 making the assumption that Poisson’s ratio for skin and other collagenous tissues is 0.5 incorrect. This may lead to errors in calculating models dependent on an exact value of this ratio . In addition, a relationship was shown to exist between the resonant frequency and the elastic modulus [5, 6]. The modulus measured using vibrational OCT and that determined from tensile incremental stress-strain curves for decellularized dermis and silicone rubber had a correlation coefficient in excess of 0.95 demonstrating that the modulus measured using vibrational OCT was very similar to the tensile modulus measured using incremental stress-strain curves [5,6]. The relationship between the modulus determined from vibrational and tensile testing is given by equation (1) where Ev and Et are the moduli determined from vibrational and tensile measurements in MPas.
Ev=1.026 Et + 0.0046 (1)
The measured value of the moduli did not depend on an assumed value of Poisson’s ratio. Results of studies on decellularized dermis and silicone rubber at frequencies of between 50 and 1000 Hz suggested that the viscous component of the modulus measured at frequencies at or above the resonant frequency was 3%-4% . Based on these results the modulus measured using vibrational OCT was considered to be an approximation of the “elastic modulus” and the viscous component was negligible . While the modulus determined from vibrational OCT is an approximation of the elastic modulus it depends on the strain since not all the collagen fibers are recruited to bear loads at low strains  (Figure 3).
Use of Vibrational OCT to Measure the Moduli of Skin in Vivo
Vibrational OCT has been used to image and measure the mechanical properties of skin and scar tissue in vivo. Results of in vivo vibrational OCT studies suggest that the modulus of normal skin is lower than that of scar tissue  and that of the margins of a healed scar have a resonant frequency different than that of normal skin (Figure 4). Recent results indicate that the margins of a scar can be mapped using vibrational OCT since the resonant frequencies and moduli seen at the interface reflect both the resonant frequency of normal skin and that of the scar tissue.
Using vibrational OCT the resonant frequency and moduli of the components of skin and scar tissue can be measured non-invasively and non-destructively. The numbers generated reflect to a first approximation the elastic moduli and do not depend on measurement of other parameters. The technique in vitro is calibrated using incremental tensile measurements and vibrational OCT results on the same sample. Using images generated by OCT, maps of the modulus as a function of position can be generated that will be useful in marking the margins of scars, tumors as well as to evaluate the effects of cosmetic treatments to the skin.
- Yamada H (1970) Strength of Biological Materials. Williams and Wilkins, Baltimore, MD.
- Fung YC (1993) Biomechanics: Mechanical Properties of Living Tissue. (2nd edn), Springer, NY.
- Dunn MG, Silver FH (1983) Viscoelastic behavior of human connective tissues: Relative contribution of viscous and elastic components. Conn Tis Res 12: 59-70.
- Silver FH, Shah R (2016) Measurement of mechanical properties of natural and engineered implants. Adv Tissue Eng Regenerative Med 1: 1-9.
- Shah R, DeVore D, Pierce MG (2016) Morphomechanics of dermis-A method for non-destructive testing of collagenous tissues. Skin Res Tech Nov 27.
- Shah R, Pierce MC, Silver FH (2017) A method for non-destructive mechanical testing of tissues and implants. J Biomed Mat Res 105A: 5-22.
- Silver FH, Freeman J, DeVore D (2001) Viscoelastic properties of human skin and processed dermis. Skin Res. Tech. 7: 18-23.
- Seehra GP, Silver FH (2006) Viscoelastic properties of acid- and alkaline-treated human dermis: A correlation between total surface charge and elastic modulus. Skin Res Tech 12: 190-198.
- Silver FH (1994) Biomaterials, Medical Devices and Tissue Engineering: An Integrated Approach, Chapman and Hall, London.
- Silver FH (2006) Mechanosensing and mechanochemical transduction in extracellular matrix. Springer, NY.
- Sherman VR, Yang W, Meyes MA (2015) The materials science of collagen. J Mech Behav Biomed Mat 52: 22-50.
- Freeman JW, Silver FH (2004) Elastic energy storage in unmineralized and mineralized extracellular matrices (ECMs): A comparison between molecular modeling and experimental measurements. J Theoretical Biology 229: 371-381.
- Dunn MG, Silver FH, Swann DA (1985) Mechanical analysis of hypertrophic scar tissue: Structural basis for apparent increased rigidity. J Invest Dermatol 84: 9-13.
- Fujimoto J, Swanson E (2016) The development, commercialization, and impact of optical coherence tomography. Invest Ophthalmol 57: 1-13.
- Shah RG, Silver FH (2017) Viscoelastic behavior of tissues and implant materials: Estimation of the elastic and viscous contribution using optical coherence tomography and vibrational analysis. J Biomed Tech Res 3: 105.
- Shah RG, Silver FH (2017) Vibrational analysis of extracellular matrix scaffolds: comparison of skin, dermis, cartilage and subchondral bone. Adv Tiss Eng Regen Med 2: 00048.
- Journal of Spine Diseases
- Journal of Cell Signaling & Damage-Associated Molecular Patterns
- International Journal of Surgery and Invasive Procedures(ISSN:2640-0820)
- Ophthalmology Clinics and Research(ISSN:2638-115X)
- Journal of Clinical Trials and Research(ISSN:2637-7373)
- Journal of Forensic Research and Criminal Investigation
- International Journal of Anaesthesia and Research(ISSN:2641-399X)